Level Ib audits
The Level Ib audit offers a more accurate measurement of the Linac output than the Level I audit. ACDS staff visit the facility and perform on-site measurements using ACDS dosimetry equipment. The Level Ib is performed for new Linac installations, or at the facility’s request.
The Level Ib audit meets requirements of ROHPG LINAC funding.
The Level Ib audit is a more accurate independent measurement than the Level I audit. The Level Ib audit determines the absorbed dose to water per monitor unit for mega-voltage photon and electron beams under reference conditions. The ACDS visits the requesting facility and performs on-site measurements in the facility’s own water tank. All other dosimetry equipment is brought in by the ACDS, ensuring an independent audit.
The Level Ib audit consists of the following modalities:
- Reference beams (photons and electrons)
- FFF beams
- Small field beams (field trial).
The Level Ib audit is not part of the scheduled program and is performed at the request of the facility. The Level Ib is typically conducted for new facilities or LINACs, or as a part of the follow up process where other audits do not fall inside the Optimal Level.
Please contact acdsarpansa.gov.au to obtain a quote for a new LINAC Level Ib audit.
The Audit results are determined by the percentage deviation of the Facility Stated Dose Output from the ACDS Determined Dose Output, for each clinical beam. All beams on the linac are defined as ‘clinical’ unless otherwise stated by the Facility. Current Result Levels are given in the following table.
|Level||% Deviation (Client Stated/ACDS Determined)|
|Action||> 1.4 and ≤ 2.1||> 2.2 and ≤ 3.3|
|Outside Tolerance||> 2.1||> 3.3|
An overall Audit Outcome for the linac is determined, which is equal to the lowest result recorded for an individual ‘clinical’ beam.